Titan is worldwide. Our head office is located at 4130 93 ST NW, Edmonton, AB T6E 5P5Q
Titan Logix warranty periods:
- Twelve (12) months for all products manufactured by Titan Logix.
- Twenty-four (24) months for Hastelloy probe (Application data sheet must be completed and accepted by Titan Logix)
- Thirty-six (36) months for TD100 transmitter.
The warranty periods shall be calculated from the date of shipment by Seller. This warranty does not cover those damages incurred due to corrosion of the wetted parts.
Probe failure from corrosion is not covered by this warranty. Buyer must follow the proper maintenance and cleaning protocols of the probe as outlined in the operating manual.
Warranty on repair products is twelve (12) months from date of repair. Warranty is limited only to the parts repaired.
This means that the input voltage to the display is below 8 volts. Look for corroded wire connections where the power cable connects. Normally this entails taking the unit in for service.
This means that the HH alarm level is set too close to the Spill alarm. This usually happens when an offset calibration has moved the level too far down and the display locks and will not move anymore. Reprogramming the TD transmitter is normally the correction for this problem.
Insert keywords here
This usually means that the TD transmitter is loose on the probe. Tighten the transmitter so the unit will not turn by hand. Do not overtighten.
¿Aún tiene preguntas?
¿No encuentra la respuesta que busca? Póngase en contacto con nuestro equipo de Atención al Cliente y un miembro de nuestro personal estará le ayudará con mucho gusto.